//Generated ALM file 000011_113c_06.alm.ss For Sourcesafe
//Byte, 0, Bit, Alarm Desc, Category
1	0	7	@C491 Current output X simulation
 Remedy information
Deactivate simulation@	@General@
1	0	6	@M168 Coating detected
 Remedy information
Clean measuring tube@	@General@
1	0	5	@C431 Trim X
 Remedy information
Carry out trim@	@General@
2	0	8	@842 Process limit
 Remedy information
Low flow cut off active!
1. Check low flow cut off configuration@	@General@
2	0	6	@C453 Flow override
 Remedy information
Deactivate flow override@	@General@
2	0	2	@C485 Measured variable simulation 
 Remedy information
Deactivate simulation@	@General@
3	0	8	@M438 Dataset
 Remedy information
1. Check data set file
2. Check device configuration
3. Up- and download new configuration@	@General@
3	0	7	@F437 Configuration incompatible
 Remedy information
Restart device@	@General@
3	0	6	@F262 Sensor electronic connection faulty
 Remedy information
1. Check or replace connection cable between sensor electronic module (ISEM) and main electronics
2. Check or replace ISEM or main electronics@	@General@
3	0	4	@F242 Software incompatible
 Remedy information
1. Check software
2. Flash or change main electronics module@	@General@
3	0	3	@F083 Memory content
 Remedy information
1. Restart device
2. Restore HistoROM S-DAT backup ('Device reset' parameter)
3. Replace HistoROM S-DAT@	@General@
3	0	2	@F382 Data storage
 Remedy information
1. Insert T-DAT
2. Replace T-DAT@	@General@
4	0	8	@441 Current output X
 Remedy information
1. Check process
2. Check current output settings@	@General@
4	0	7	@F275 I/O module X defective
 Remedy information
Change I/O module@	@General@
4	0	6	@F273 Main electronic failure
 Remedy information
Change electronic@	@General@
4	0	5	@F410 Data transfer
 Remedy information
1. Check connection
2. Retry data transfer@	@General@
4	0	4	@M311 Electronic failure
 Remedy information
1. Do not reset device
2. Contact service@	@General@
4	0	3	@F276 I/O module X faulty
 Remedy information
1. Restart device
2. Change I/O module@	@General@
4	0	2	@F803 Current loop @1
 Remedy information
1. Check wiring
2. Change I/O module@	@General@
4	0	1	@C484 Failure mode simulation
 Remedy information
Deactivate simulation@	@General@
5	0	8	@C502 CT activation/deactivation failed
 Remedy information
Follow the sequence of the custody transfer activation/deactivation: First authorized user login, then set the DIP switch on the main electonic module@	@General@
5	0	7	@C495 Diagnostic event simulation
 Remedy information
Deactivate simulation@	@General@
5	0	5	@C494 Switch output simulation X
 Remedy information
Deactivate simulation switch output@	@General@
5	0	4	@C493 Simulation pulse output X
 Remedy information
Deactivate simulation pulse output@	@General@
5	0	3	@C492 Simulation frequency output X
 Remedy information
Deactivate simulation frequency output@	@General@
5	0	2	@443 Pulse output X
 Remedy information
1. Check process
2. Check pulse output settings@	@General@
5	0	1	@442 Frequency output X
 Remedy information
1. Check process
2. Check frequency output settings@	@General@
6	0	7	@C412 Processing download
 Remedy information
Download active, please wait@	@General@
6	0	5	@F082 Data storage
 Remedy information
1. Check module connections
2. Change electronic modules@	@General@
6	0	4	@444 Current input X
 Remedy information
1. Check process
2. Check current input settings@	@General@
6	0	3	@C486 Current input X simulation
 Remedy information
Deactivate simulation@	@General@
6	0	1	@F882 Input signal
 Remedy information
1. Check input configuration
2. Check external device or process conditions@	@General@
7	0	6	@Function Check@	@General@
7	0	5	@Out of Specification@	@General@
7	0	4	@Failure@	@General@
7	0	3	@Critical Power Failure@	@General@
7	0	2	@Device variable alert@	@General@
7	0	1	@Maintenance required@	@General@
9	0	7	@Electronic failure@	@General@
9	0	6	@Environmental conditions out of range@	@General@
9	0	5	@Voltage conditions out of range@	@General@
9	0	4	@Watchdog reset executed@	@General@
9	0	3	@Volatile memory error@	@General@
9	0	2	@Non-Volatile memory failure@	@General@
9	0	1	@Simulation active@	@General@
10	0	1	@Reserved@	@General@
11	0	4	@Quinary Analog Channel Saturated@	@General@
11	0	3	@Quaternary Analog Channel Saturated@	@General@
11	0	2	@Tertiary Analog Channel Saturated@	@General@
11	0	1	@Secondary Analog Channel Saturated@	@General@
12	0	2	@Duplicate master detected@	@General@
12	0	1	@Subdevice list changed@	@General@
13	0	4	@Block transfer pending@	@General@
13	0	3	@Bandwidth allocation pending@	@General@
13	0	1	@Capacity Denied@	@General@
14	0	4	@Quinary Analog Channel Fixed@	@General@
14	0	3	@Quaternary Analog Channel Fixed@	@General@
14	0	2	@Tertiary Analog Channel Fixed@	@General@
14	0	1	@Secondary Analog Channel Fixed@	@General@
15	0	7	@835 Process temperature too low
 Remedy information
Increase process temperature@	@General@
15	0	6	@834 Process temperature too high
 Remedy information
Reduce process temperature@	@General@
15	0	5	@833 Electronic temperature too low
 Remedy information
Increase ambient temperature@	@General@
15	0	4	@832 Electronic temperature too high
 Remedy information
Reduce ambient temperature@	@General@
16	0	8	@F201 Device failure
 Remedy information
Restart device@	@General@
16	0	7	@F272 Main electronic failure
 Remedy information
Restart device@	@General@
16	0	6	@F271 Main electronic failure
 Remedy information
1. Restart device
2. Change main electronic module@	@General@
16	0	5	@F270 Main electronic failure
 Remedy information
Change main electronic module@	@General@
16	0	4	@F252 Modules incompatible
 Remedy information
1. Check electronic modules
2. Check if correct modules are available (e.g. NEx, Ex)
3. Replace electronic modules@	@General@
17	0	6	@531 Empty pipe adjustment faulty  
 Remedy information
Execute EPD adjustment@	@General@
17	0	4	@F512 Sensor electronic (ISEM)  faulty
 Remedy information
1. Check ECC recovery time
2. Turn off ECC@	@General@
17	0	3	@961 Electrode potential out of specification
 Remedy information
1. Check process conditions
2. Check ambient conditions@	@General@
17	0	2	@937 Sensor symmetry
 Remedy information
1. Eliminate external magnetic field near sensor 
2. Turn off diagnostic message@	@General@
18	0	7	@F537 Configuration
 Remedy information
1. Check IP addresses in network 2. Change IP address@	@General@
18	0	5	@F181 Sensor connection
 Remedy information
1. Check sensor cable and sensor
2. Execute Heartbeat Verification
3. Replace sensor cable or sensor@	@General@
18	0	3	@043 Sensor short circuit
 Remedy information
1. Check sensor cable and sensor
2. Execute Heartbeat Verification
3. Replace sensor cable or sensor@	@General@
18	0	2	@F383 Memory content
 Remedy information
1. Restart device
2. Delete T-DAT via  'Reset device' parameter
3. Replace T-DAT@	@General@
18	0	1	@F283 Memory content
 Remedy information
1. Reset device 2. Contact service@	@General@
19	0	5	@C530 Electrode cleaning is running
 Remedy information
Turn off ECC@	@General@
19	0	2	@F375 I/O-X communication failed
 Remedy information
1. Restart device
2. Check if failure recurs
3. Replace module rack inclusive electronic modules@	@General@
19	0	1	@C302 Device verification active
 Remedy information
Device verification active, please wait.@	@General@
20	0	8	@F378 Supply voltage ISEM faulty
 Remedy information
Check supply voltage to the ISEM@	@General@
20	0	5	@377 Sensor electronic (ISEM)  faulty
 Remedy information
1. Activate empty pipe detection
2. Check partial filled pipe and installation direction
3. Check sensor cabling 
4. Deactivate diagnostic 377@	@General@
20	0	2	@F540 Custody transfer mode failed
 Remedy information
1. Power off device and toggle DIP switch
2. Deactivate custody transfer mode 
3. Reactivate custody transfer mode
4. Check electronic components@	@General@
21	0	2	@938 EMC interference
 Remedy information
1. Check ambient conditions regarding EMC influence 
2. Turn off diagnostic message@	@General@
22	0	8	@F599 Custody transfer logbook full
 Remedy information
1. Deactivate custody transfer mode
2. Clear custody transfer logbook (all 30 entries)
3. Activate custody transfer mode@	@General@
22	0	7	@F387 HistoROM data faulty
 Remedy information
Contact service organization@	@General@
22	0	6	@C594 Relay output @1 simulation
 Remedy information
Deactivate simulation switch output@	@General@
22	0	5	@M169 Conductivity measurement failed
 Remedy information
1. Check grounding conditions
2. Deactivate conductivity measurement@	@General@
22	0	3	@F520 I/O X hardware configuration invalid
 Remedy information
1. Check I/O hardware configuration
2. Replace wrong I/O module
3. Plug the module of double pulse output on correct slot@	@General@
22	0	2	@F332 Writing in HistoROM backup failed
 Remedy information
Replace user interface board
Ex d/XP: replace transmitter@	@General@
22	0	1	@M303 I/O X configuration changed
 Remedy information
1. Apply I/O module configuration (parameter 'Apply I/O configuration')
2. Afterwards reload device description and check wiring@	@General@
23	0	7	@F373 Sensor electronic (ISEM)  faulty
 Remedy information
Transfer data or reset device@	@General@
23	0	6	@F372 Sensor electronic (ISEM)  faulty
 Remedy information
1. Restart device
2. Check if failure recurs
3. Replace sensor electronic module (ISEM)@	@General@
23	0	5	@F180 Temperature sensor defective
 Remedy information
1. Check sensor connections
2. Replace sensor cable or sensor
3. Turn off temperature measurement@	@General@
23	0	2	@M330 Flash file invalid
 Remedy information
1. Update firmware of device
2. Restart device@	@General@
23	0	1	@F331 Firmware update failed
 Remedy information
1. Update firmware of device
2. Restart device@	@General@
24	0	8	@C511 ISEM settings faulty 
 Remedy information
1. Check measuring period and integration time
2. Check sensor properties@	@General@
24	0	7	@376 Sensor electronic (ISEM)  faulty
 Remedy information
1. Replace sensor electronic module (ISEM)
2. Turn off diagnostic message@	@General@
24	0	6	@962 Pipe empty
 Remedy information
1. Perform full pipe adjustment
2. Perform empty pipe adjustment
3. Turn off empty pipe detection@	@General@
24	0	5	@F170 Coil resistance
 Remedy information
Check ambient and process temperature@	@General@
24	0	4	@C496 Status input simulation @1
 Remedy information
Deactivate simulation status input@	@General@
24	0	3	@F361 I/O module X faulty
 Remedy information
1. Restart device
2. Check electronic modules
3. Change I/O Modul or main electronics@	@General@
24	0	2	@543 Double pulse output
 Remedy information
1. Check process
2. Check pulse output settings@	@General@
24	0	1	@C593 Double pulse output @1 simulation
 Remedy information
Deactivate simulation pulse output@	@General@
25	0	8	@Fld dev stat24-8@	@General@
25	0	7	@Fld dev stat24-7@	@General@
25	0	6	@Fld dev stat24-6@	@General@
25	0	5	@Fld dev stat24-5@	@General@
25	0	4	@Fld dev stat24-4@	@General@
25	0	3	@Fld dev stat24-3@	@General@
25	0	2	@Fld dev stat24-2@	@General@
25	0	1	@Fld dev stat24-1@	@General@
27	0	8	@Field Device Malfunction@	@Overview@
27	0	7	@Configuration Change@	@Overview@
27	0	6	@Cold Start@	@Overview@
27	0	5	@More Status Available@	@Overview@
27	0	4	@Loop Current Fixed@	@Overview@
27	0	3	@Loop Current Saturated@	@Overview@
27	0	2	@NPV Out of Limits@	@Overview@
27	0	1	@PV Out of Limits@	@Overview@
28	0	2	@Device Mfg/Type/Rev Mismatch@	@DeltaV@
28	0	1	@Analog/Digital Disparity Detected@	@DeltaV@
30	0	8	@Proof Test Overdue@	@DeltaV@
30	0	7	@Proof Test Due Soon@	@DeltaV@
30	0	3	@External Advise Condition@	@DeltaV@
30	0	2	@External Maint Condition@	@DeltaV@
30	0	1	@External Failed Condition@	@DeltaV@
